Continuous controllable balloon dilation: a novel approach for cervix dilation
© Arsenijevic et al.; licensee BioMed Central Ltd. 2012
Received: 28 May 2012
Accepted: 8 October 2012
Published: 22 October 2012
Cervical dilation using mechanical dilators is associated with various complications, such as uterine perforation, cervical laceration, infections and intraperitoneal hemorrhage. To achieve safe and painless cervical dilation, we constructed a new medical device to achieve confident mechanical cervical dilation: a continuous controllable balloon dilator (CCBD).
Controlled pumping of incompressible fluid into the CCBD increases the pressure and outer diameter of the CCBD, continuously dilating the cervical canal. The reliability of the CCBD was confirmed in vitro (testing for consistency and endurance, with no detected risk for breakage) and in vivo. A multi-center clinical study was conducted,with 120 pregnant women randomly assigned to one of three groups: Group I,control group, no dilation;Group II,mechanical dilation, Hegar dilator (HeD); and Group III,CCBD. The tissue material for histological evaluation was obtained from the endocervical mucosa before and after dilation using the HeD or CCBD.
The CCBD dilations were successful and had no complications in all 40 patients of Group III. The cervical tissue was markedly less damaged after CCBD dilation compared with HeD dilation (epithelium damage: 95% (HeD) vs. 45% (CCBD), P <0.001; basal membrane damage: 82.5% (HeD) vs. 27.5% (CCBD), P <0.001; stromal damage: 62.5% (HeD) vs. 37.5% (CCBD), P <0.01). Cervical hemorrhagia was observed in 90% of the patients after HeD dilation versus in 32.5% of the patients after CCBD dilation.
The CCBD should be used as a replacement for mechanical dilators to prevent uterine and cervical injury during cervical dilation.
KeywordsMedical device Balloon dilation Cervix
Although cervical dilation is most commonly reserved for childbirth, its use has expanded to a large number of diagnostic procedures (dilation and curettage for diagnosing endometrial cancer, endometrial biopsies, evaluating the causes of infertility and hysteroscopy) and therapeutic procedures (cervical stenosis, dysfunctional uterine bleeding and dysmenorrhea, inserting intrauterine devices and draining the uterine cavity) [1–3]. Mechanical instruments, such as the Hern, Hegar, Pratt, Hanks and Denniston dilators, are used to sequentially dilate the cervix [4, 5] by incrementally increasing the diameter of the inserted dilator until the dilation procedure is complete. However, these mechanical dilators require the use of appropriate force, which could permanently damage the cervical tissue and have adverse long-term effects on fertility [6–9]. Their use is associated with various potential complications, such as uterine perforation, cervical laceration, infections and intraperitoneal hemorrhages [4, 5, 10].
Several lines of evidence have suggested that cervical priming, accomplished by pharmacological agents (prostaglandin analogs), can prevent cervical laceration by reducing the force required for cervical dilation [3, 10]. The main disadvantage of pharmacological agents is that the woman might experience bleeding and cramping prior to the surgical procedure [11–13].
To achieve safe and painless cervical dilation, we constructed a medical device for reliable mechanical cervical dilation: a continuous controllable balloon dilator (CCBD). The CCBD is based on a patented solution  that combines all of the advantages of traditional mechanical devices with several new features that give the provider full process control in all of the phases of cervical dilation. We conducted a pilot study that showed a significant reduction in all of the side effects related to dilator use for the CCBD compared with metal mechanical devices.
The study (ISRCTN54007498) was conducted at the Gynecology & Obstetrics Clinics at Kragujevac Clinical Center, Serbia, and Podgorica Clinical Center, Montenegro. The data were collected by the study coordinators at the participating centers. The authors vouch for the completeness and accuracy of the data and analyses. An independent data and safety monitoring board monitored the study and reviewed the protocol compliance and outcome data. The protocol was approved by each participating center's institutional review board.
The patients were enrolled in the study using the following criteria: age between 19 and 40; pregnancy verified by an ultrasound; singleton pregnancy; gestational age ≤10; uterus and cervix with normal findings; and absence of uterine contractions or bleeding.
Patients were excluded from the study if any of the following criteria were met: any signs of spontaneous abortion; any previous attempt at an abortion or use of substances for cervical maturation; multiple pregnancy; the presence or, at minimum, the suspicion of a septic abortion, followed by an elevated body temperature of 38°C or higher, uterine pain and odorous vaginal secretions; the presence of any previous intervention performed on the uterine cervix; uterine or cervical anomalies; an intrauterine device in situ; or hemorrhagic and/or chronic diseases.
The study included 120 pregnant women randomly assigned to one of the following three groups: Group I (40 pregnant women), control group with no dilations; Group II (40 pregnant women), dilations performed using Hegar dilators (HeDs); and Group III (40 pregnant women), dilations performed using the CCBD.
The experiments were undertaken with the understanding and appropriate informed consent of each patient.
Continuous controllable balloon dilator: main characteristics
Dilation using the CCBD is performed continuously, with only one dilator placement. In this study, the CCBD was integrated into a system that enables real-time data acquisition and the monitoring of the parameters relevant to the biophysics of dilation (Figure 1A). Dilation dynamics directly depend on the flow of an incompressible fluid into the BD, which is an easily controllable parameter. Because it is an incompressible working fluid, distilled water was used with the addition of nonionic contrast medium (Ultravist-300; Schering AG, Berlin, Germany), which enabled the visual monitoring of dilation using a digital subtraction apparatus for angiography (Figure 1A).
Incompressible fluid from the hydraulic cylinder is pumped to the BD via a flexible hose (hose) and one-way valve (Figure 1A). Fluid flow is controlled by an electric motor that, via a spiral spindles/nut system, provides constant speed of the hydraulic cylinder piston and consequently constant fluid flow. The actual position of the hydraulic cylinder piston and the fluid pressure are monitored by displacement and pressure transducers. The unit for measurement and control collects signals from those two transducers. The pressure gauge is used for visual control of the current fluid pressure, while the pressure reduction valve represents a safety element. After finalization of the dilatation procedure, the choke valve is opened by remote command, which results in draining of fluid from the BD. Then the BD could be easily extracted from the cervical canal. Dilatation duration and maximum pressure in the BD represents parameters assigned by a PC-controlled unit for measurement and control. Those parameters could be monitored in real-time parameters by PC and appropriate acquisition software, which enable full controllability of the cervical canal dilatation process (Figure 1A).
Tissue material for the histological evaluation of cervical damage was obtained from the endocervical mucosa by single curettage (Novac Curette, CooperSurgical, Trumbull, Connecticut, USA) before and after dilation using the HeD or CCBD. The samples were stained with H&E and blinded analysis was performed using a light microscope (Olympus BX 51; Olympus, Hamburg, Germany). The surface areas of the hemorrhagic regions were measured according to a previously published protocol .
All of the statistical analyses were conducted using SPSS 13.0 for Windows (IBM, New York, New York, USA). The results were analyzed using Student’s t test. All of the data in this study are expressed as the mean ± standard error. P < 0.05 was considered significant.
Continuous controllable balloon dilation results
The CCBD dilations were successfully performed on all patients. The providers and patients reported no complications or problems with the CCBD, a new dilation technique. The average duration of CCBD dilation was approximately 25.5 seconds (Figure 1F) and was independent of the individual patient's cervical resistance (Figure 1F). The maximal recorded pressure in the BD during dilation was significantly lower than the test pressure for BD consistency and endurance.
Comparative analysis of biophysical phenomena during cervical dilation by HeD and CCBD
In contrast, the CCBD dilation procedure involves inserting the BD in its initial form (diameter D i ≈ 4.5 mm) into the cervical canal, which results in a very low resistance to penetration (Figure 2B). Pumping incompressible fluid into the BD increases the outer diameter and pressure of the BD, which dilates the cervical canal. Dilation is performed synchronously along the entire length L of the cervical canal, where the relative movement between the tissue/BD contact pair is reduced to almost zero. In this case, the cervical tissue also opposes changes in its geometry, which is characterized by a much more uniform pressure distribution p(x) (Figure 2B). All of the components of the involved forces participate in dilation, which significantly reduces any side effects.
Cervical tissue is markedly less damaged and the extent of cervical hemorrhagia is significantly lower after CCBD compared with HeD dilation
Based on our results, CCBD dilation is a novel, non-invasive, fully controllable and safe clinical procedure; and compared with the predominant mechanical method for cervical dilation (HeD) , CCBD is a less invasive and more reliable system for cervical dilation.
The main factors leading to uterine and cervical injury during dilation procedures that use metal mechanical dilators are the provider's inexperience, an abnormal uterine cavity and/or an anteflexed or retroflexed uterus [13, 16–19]. In addition, adolescents have a higher risk for cervical laceration because they usually have small, physiologically immature cervices that are difficult to dilate [20–22]. Another significant problem with current cervical dilators is a lack of control over the rate of dilation – extremely rapid cervical dilation often leads to cervical laceration [23, 24].
All of the aforementioned risk factors (the provider's inexperience, the lack of control over the rate of dilation, differences in uterine position and cervical immaturity) could be overcome using the CCBD. Dilation with the CCBD is fully controllable, safe and independent of the provider's experience. In the case of incompressible fluid BD breakage (which did not happen during our study), the leakage of only a few drops of fluid results in a sudden pressure drop in the BD; therefore, the risk for patient injury in this case is almost non-existent. Both the trainees and the experienced providers could use the CCBD without the risk for cervical laceration or uterine perforation. It is noteworthy that CCBD cervical dilation was successfully performed on all of the patients, regardless of the patient's age, cervical stenosis, uterine position and previous obstetric history. In addition, no side effects were observed during the procedure or after successful cervical dilation. However, it is important to note that due to limited number of patients enrolled in this study, a large prospective study should be conducted in order to confirm the effectiveness of CCBD dilation.
We suggest the CCBD as a replacement for mechanical dilators, with the main purpose of preventing uterine and cervical injury during cervical dilation. Depending on the opinion of the physician, the CCBD could be used alone or in combination with cervical priming agents.
In the end, it should be emphasized that CCBD dilation is a new, original, non-invasive, fully controllable and safe technique for cervical dilation. However, because of the limited number of patients enrolled in this study, a large prospective study should be conducted in the future to further confirm the effectiveness of CCBD dilation.
Institutional approval for the study was granted by the Ethics Committee of the Kragujevac Clinical Center and the Ethics Committee of the Podgorica Clinical Center in accordance with internationally accepted ethical standards (the Helsinki Declaration of 1964, last revised in 2005). The experiments were undertaken with the understanding and appropriate informed consent of each patient.
Continuous controllable balloon dilator
Hematoxylin and eosin
The authors are thankful to Mr Milan Milojevic for excellent technical assistance.This study was financed by grants from Ministry of Science Republic of Serbia and Faculty of Medicine University of Kragujevac.
- Newmann S, Dalve-Endres A, Drey EA: Cervical preparation for surgical abortion from 20 to 24 weeks' gestation. Contraception. 2008, 77: 308-314. 10.1016/j.contraception.2008.01.004.View ArticlePubMedGoogle Scholar
- Fox MC, Hayes JL: Cervical preparation for second-trimester surgical abortion prior to 20 weeks of gestation. Contraception. 2007, 76: 486-495.View ArticlePubMedGoogle Scholar
- Finer LB, Henshaw SK: Abortion incidence and services in the united states in 2000. Perspect Sex Reprod Health. 2003, 35: 6-15. 10.1363/3500603.View ArticlePubMedGoogle Scholar
- Gelber S, Sciscione A: Mechanical methods of cervical ripening and labor induction. Clin Obstet Gynecol. 2006, 49: 642-657. 10.1097/00003081-200609000-00022.View ArticlePubMedGoogle Scholar
- Biron-Shental T, Fishman A, Fejgin MD: Medical and mechanical methods for cervical ripening. Int J Gynaecol Obstet. 2004, 85: 159-160. 10.1016/j.ijgo.2003.08.006.View ArticlePubMedGoogle Scholar
- Kloeck FK, Jung H: In vitro release of prostaglandins from the human myometrium under the influence of stretching. Am J Obstet Gynecol. 1973, 115: 1066-1069.View ArticlePubMedGoogle Scholar
- Hulka JF, Lefler HT, Anglone A, Lachenbruch PA: A new electronic force monitor to measure factors influencing cervical dilation for vacuum curettage. Am J Obstet Gynecol. 1974, 120: 166-173.View ArticlePubMedGoogle Scholar
- Golditch IM, Glasser MH: The use of laminaria tents for cervical dilation prior to vacuum aspiration abortion. Am J Obstet Gynecol. 1974, 119: 481-485.View ArticlePubMedGoogle Scholar
- Gupta JK, Johnson N: Should we use prostaglandins, tents or progesterone antagonists for cervical ripening before first trimester abortion?. Contraception. 1992, 46: 489-497. 10.1016/0010-7824(92)90152-J.View ArticlePubMedGoogle Scholar
- Lichtenberg ES, Paul M, Jones H: First trimester surgical abortion practices: a survey of national abortion federation members. Contraception. 2001, 64: 345-352. 10.1016/S0010-7824(01)00279-7.View ArticlePubMedGoogle Scholar
- Fiala C, Gemzell-Danielsson K, Tang OS, von Hertzen H: Cervical priming with misoprostol prior to transcervical procedures. Int J Gynaecol Obstet. 2007, 99: 168-171.View ArticleGoogle Scholar
- Gómez Ponce de León R, Wing DA: Misoprostol for termination of pregnancy with intrauterine fetal demise in the second and third trimester of pregnancy – a systematic review. Contraception. 2009, 79: 259-271. 10.1016/j.contraception.2008.10.009.View ArticlePubMedGoogle Scholar
- Allen RH, Goldberg AB: Cervical dilation before first-trimester surgical abortion (<14 weeks' gestation). SFP Guideline 20071. Contraception. 2007, 76: 139-156.View ArticlePubMedGoogle Scholar
- Arsenijević S, Cakic N: Instrument for fluid injection and dilation probe for implantation in body cavities. European Patent No.1299146;27 October 2004. Available at http://www.google.com/patents/EP1299146B1, published in Patent office journal, 2006, Vol. 2044, page 943.
- Volarevic V, Milovanovic M, Arsenijevic N, Lukic M: The new semi-quantitative method for determination of liver damage after concanavalin a administration. Ser J Exp Clin Res. 2010, 11: 45-48.Google Scholar
- Child TJ, Thomas J, Rees M, MacKenzie IZ: Morbidity of first trimester aspiration termination and the seniority of the surgeon. Hum Reprod. 2001, 16: 875-878. 10.1093/humrep/16.5.875.View ArticlePubMedGoogle Scholar
- Kaunitz AM, Rovira EZ, Grimes DA, Schulz KF: Abortions that fail. Obstet Gynecol. 1985, 66: 533-537.PubMedGoogle Scholar
- Ferris LE, McMain-Klein M, Colodny N, Fellows GF, Lamont J: Factors associated with immediate abortion complications. Can Med Assoc J. 1996, 154: 1677-1685.Google Scholar
- Fisher J, Anthony GS, McManus TJ, Coutts JRT, Calder AA: Use of a force measuring instrument during cervical dilation. J Med Eng Technol. 1981, 4: 194-195.View ArticleGoogle Scholar
- Schulz KF, Grimes DA, Cates W: Measures to prevent cervical injury during suction curettage abortion. Lancet. 1983, 1: 1182-1185.View ArticlePubMedGoogle Scholar
- Cates JW, Schulz KF, Grimes DA: The risks associated with teenage abortion. N Engl J Med. 1983, 309: 621-624. 10.1056/NEJM198309153091101.View ArticlePubMedGoogle Scholar
- Jain JK, Mishell DR: A comparison of intravaginal misoprostol with prostaglandin E2 for termination of second-trimester pregnancy. N Engl J Med. 1994, 331: 290-293. 10.1056/NEJM199408043310502.View ArticlePubMedGoogle Scholar
- Hern WM: Laminaria versus dilapan osmotic cervical dilators for outpatient dilation and evacuation abortion: randomized cohort comparison of 1001 patients. Am J Obstet Gynecol. 1994, 171: 1324-1328.View ArticlePubMedGoogle Scholar
- Hayes JL, Fox MC: Cervical dilation in second-trimester abortion. Clin Obstet Gynecol. 2009, 52: 171-178. 10.1097/GRF.0b013e3181a2b3cd.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.