Evaluation of surgically assisted rapid maxillary expansion with piezosurgery versus oscillating saw and chisel osteotomy - a randomized prospective trial
© Rana et al.; licensee BioMed Central Ltd. 2013
Received: 4 August 2012
Accepted: 30 January 2013
Published: 17 February 2013
Ultrasonic bone-cutting surgery has been introduced as a feasible alternative to the conventional sharp instruments used in craniomaxillofacial surgery because of its precision and safety. The piezosurgery medical device allows the efficient cutting of mineralized tissues with minimal trauma to soft tissues. Piezoelectric osteotome has found its role in surgically assisted rapid maxillary expansion (SARME), a procedure well established to correct transverse maxillary discrepancies. The advantages include minimal risk to critical anatomic structures. The purpose of this clinical comparative study (CIS 2007-237-M) was to present the advantages of the piezoelectric cut as a minimally invasive device in surgically assisted, rapid maxillary expansion by protecting the maxillary sinus mucosal lining.
Thirty patients (18 females and 12 males) at the age of 18 to 54 underwent a surgically assisted palatal expansion of the maxilla with a combined orthodontic and surgical approach. The patients were randomly divided into two separate treatment groups. While Group 1 received conventional surgery using an oscillating saw, Group 2 was treated with piezosurgery. The following parameters were examined: blood pressure, blood values, required medication, bleeding level in the maxillary sinus, duration of inpatient stay, duration of surgery and height of body temperature.
The results displayed no statistically significant differences between the two groups regarding laboratory blood values and inpatient stay. The duration of surgery revealed a significant discrepancy. Deploying piezosurgery took the surgeon an average of 10 minutes longer than working with a conventional-saw technique. However, the observation of the bleeding level in the paranasal sinus presented a major and statistically significant advantage of piezosurgery: on average the bleeding level was one category above the one of the remaining patients.
This method of piezoelectric surgery with all its advantages is going to replace many conventional operating procedures in oral and maxillofacial surgery.
KeywordsMaxillary expansion Piezosurgery Saw Neurologic Pain
Transverse expansion of the maxilla was first done in 1860 by means of an orthodontic appliance. In the following decennia, the orthodontic treatment evolved. The theory of distraction was first published in 1905 by Codivilla . The combined surgical and orthodontic treatment for maxillary expansion was introduced in 1938 for skeletally matured patients. The first successful use of distraction on the femur of a significant group of patients was published in 1990 . In 1999, the first bone-borne distractor was introduced .
Once skeletal maturity has been reached, orthodontic treatment alone cannot provide a stable widening of the constricted maxilla in cases of deficiencies of more than 5 mm. In general, an orthodontist can camouflage transverse discrepancies less than 5 mm with orthopedic forces alone . Tooth extractions for alignment of dental arches are often unnecessary . As mentioned before, surgically assisted rapid maxillary expansion (SARME) is a form of distraction that was applied before its biological healing principles were known . Physicians have to decide between two methods of expansion: SME (slow maxillary expansion) and RME (rapid maxillary expansion). Applying SME, the maxilla is broadened by 0.5 to 1 mm per week; meanwhile, using RME demands an expansion of 0.6 to 0.8 mm three times a day. Both methods have advantages and disadvantages . The surgically assisted rapid maxillary expansion is a method which, using combined orthodontic-oral surgical treatment, leads to a distinctive extension of the midline palatal suture. Thereby, it is possible to avoid extractions, widen the nasal floor and support the change from oral to nasal breathing .
SARME is considered a procedure with little risk of serious complications; however, several complications are mentioned in the literature, varying from life threatening epistaxis to a cerebrovascular accident, skull base fracture with reversible oculomotor nerve pareses and orbital compartment syndrome [8–10]. Less serious complications reported are postoperative hemorrhage, pain, sinusitis, palatal tissue irritation/ulceration, asymmetrical expansion, nasal septum deviation, periodontal problems and relapse .
In 1976 Bell and Epker, as well as Neubert in 1989, described the surgically assisted maxillary expansion, all using an oscillating saw that injured the mucous membrane of the maxillary sinus . Unfortunately, only a little information exists on how to preserve this mucous sinus membrane during the intervention. However, in 2001 Vercelotti described a new technique in osseous surgery which overcame the limits of traditional instrumentation in oral bone surgery by modifying and improving conventional ultrasound technology. Therefore, Vercelotti is known as the inventor of piezosurgery [13, 14], a technique that allows the soft tissue to rest and a tendency for less bleeding [15–17]. It transmits a special modular ultrasonic vibration frequency on the scalpel. Not only is this technique clinically effective, but histological and histomorphometric evidence of wound healing and bone formation in experimental animal models has shown that tissue response is more favorable in piezosurgery than it is in conventional bone-cutting techniques, such as with diamond or carbide rotary instruments . Voltage applied to a polarized piezoceramic causes it to expand in the direction of and contract perpendicular to polarity. A frequency of 25 to 29 kHz is used because the micromovements that are created at this frequency (ranging between 60 and 210 μm) cut only mineralized tissue; neurovascular tissue and other soft tissue is cut at frequencies higher than 50 kHz [19–22].
Piezoelectric devices are an innovative ultrasonic technique for safe and effective osteotomy or osteoplasty compared with traditional hard and soft tissue methods that use rotating instruments because of the absence of macrovibrations, ease of use and control, and safer cutting, particularly in complex anatomical areas. Its physical and mechanical properties have several clinical advantages: precise cutting, sparing of vital neurovascular bundles and better visualization of the surgical field. Piezoelectric bone surgery seems to be more efficient in the first phases of bony healing; it induces an earlier increase in bone morphogenetic proteins, controls the inflammatory process better, and stimulates remodeling of bone as early as 56 days after treatment .
SARME is reported to be performed under either general or local anesthesia, but with differences in surgical technique. Pterygo-maxillary separation is not recommended by those who have performed SARME under local anesthesia, as it is performed blindly and can produce profuse bleeding from the descending palatine bundle that is not easily controlled without a maxillary down-fracture [24, 25]. Separation of the pterygoid junction is thus particularly useful if greater posterior expansion is desired . The use of ultrasonic vibrations for fracture of the pterygoid plates during orthognathic surgery has been recently reported by Ueki et al. . In other words, the scope of the study is to answer the question of whether a surgically assisted rapid maxillary expansion using Piezosurgery® (Mectron, Carasco, Italy) without using saws and chisels is as effective as one applying conventional procedures.
Approval for the study was obtained from the relevant ethics committee at the University of Muenster, Germany (CIS 2007-237-M). In addition, positive written consent was obtained from each subject who participated in the study.
A total of 30 consecutive adult patients (12 males and 18 females) with an indication for surgically assisted maxillary expansion were prospectively and observer-blind enrolled. The division into two groups occurred randomly. To reduce bleeding in the maxillary sinus, patients of the first group were treated conventionally with an oscillating saw, while patients of the second group were treated with a piezoelectrical saw. Exclusion criteria were syndromes like cleft or craniofacial deformities. Patients with systemic disorders, cardiac diseases, diabetes mellitus, epilepsy, infectious diseases, coagulative disorders, pregnancy and patients receiving any regular drug therapy (for example, antiphlogistic), except oral contraceptives, were excluded from the study.
Piezosurgical medical device
Consort flow diagram
At the time of presentation, 48 patients were assessed for eligibility to be included in the study. Out of these, 20% of the patients (n = 10) were not included in the study as 12% of patients (n = 6) did not meet the inclusion criteria while 8% (n = 4) did not want to participate in the study. A total of 38 patients were randomly allocated in two groups with 19 patients allocated in each group for intervention. In the group treated by an oscillating saw 100% of the patients (n = 19) received the selected intervention. In the group operated on with the piezoelectric device 100% patients (n = 19) received the selected intervention. Among the 19 patients who were operated on using an oscillating saw, 15% (n = 3) were lost to follow-up as these patients came from far areas and could not travel due to economic or personal reasons. Of the 19 patients who were managed using the piezoelectric device, 5% (n = 1) were lost to follow-up. The 16 patients who received treatment using an oscillating saw in Group 1 were available for follow-up; 1 of them had their data lost during the data analysis procedure. So the total number of patients who were analyzed for an oscillating saw was 15.
Randomization was done using a computer-based software “EpiCalc2000” (Brixton-Health, London, United Kingdom). The software was used to generate serial numbers 1 to 100 randomly into two groups and those patients who fulfilled the inclusion criteria were allocated serial numbers according to date and sequence of admission to the hospital. The person responsible for conducting the measurements at the time of assessment of variables was blinded regarding the type of procedure that was conducted.
The sample size for the study was planned using the data of a pretest (n = 10). In this pretest, the alpha error level was set as 0.05 with 95% confidence level and a study power of 80%. The hematosinus of patients after definitive surgical assisted maxillary expansion of hard and soft tissue in orthgnathic patients was 11.9 + 5.8 mm in Group 1 and 5.7 ± 2.6 mm in Group 2. The calculated sample size was 15 cases in each group with an indication for surgically assisted maxillary expansion.
Blood loss and hematosinus
Blood loss and hematosinus was measured using the paranasal sinus exposures preoperative and direct postoperative. A potential hematosinus was noticed by the measurement of the filling level of the maxillary sinus.
Post-operative pain analysis
Pain analysis was performed using a visual analogue scale as described previously . Briefly, pain was graded on a scale from 0 to 10, where a 0 denotes no pain and 10 maximum intensity of pain.
Post-operative neurological score analysis
Neurological analysis was done for infraorbital nerve bilaterally as described previously  with some modifications. This method was basically created for nerve reconstructions. We used this neurological score to evaluate nerve dysfunction after orthognathic surgery. Briefly, the skin of the infraorbital region and the upper lip was checked using a cotton test for touch sensation, a pinprick test using a needle for sharp pain and a blunt instrument for pressure. In addition, a two-point discrimination test was performed on these regions. The results were graded on a score ranging from 0 to 13, where 13 was set as the worst neurological score. The scores were collected on the fourth day and at six months postoperatively.
Patient satisfaction with surgical treatment
All patients were given a questionnaire before discharge from the hospital. Patients were questioned based on the subjective perception of the comfort and satisfactory concerning the post-operative cooling therapy. The data were graded in a scale ranging from 1 to 4, with 1 set as very satisfied and 4 not satisfied.
All data are expressed as mean values ± 1 SEM (standard error of the mean). A one-way analysis of variance (ANOVA) with post hoc Bonferroni’s test for multiple comparisons of means was used for repeated measures. The Student’s t-test was applied for quantitative variables. A P- value < 0.05 was considered significant. Statistical analysis was done with SPSS software for windows Version 14.0 (SPSS Inc., Chicago, IL, USA).
Baseline characteristics of patients
Parameters (unit) ± SD
Group 1 (n = 15)
Group 2 (n = 15)
Gender male – no. /total no. (%)
27.4 ± 9.4
29.8 ± 11.3
14.63 ± 1.41
13.79 ± 0.87
42.5 ± 3.8
40.2 ± 3.5
Body core temperature (°C)
36.2 ± 0.2
36.3 ± 0.3
Blood pressure (mmHg)
118.7 ± 9.9
123.3 ± 10.5
13.75 ± 1.19
12.83 ± 0.87
3.1 ± 0.3
1.9 ± 0.2
Blood loss and hematosinus
Blood loss during surgery added up to a mean of 0.87 ± 0.68 g/dl for Group 1 and 0.95 ± 0.65 g/dl for Group 2. The blood loss during surgery with both operative techniques remained without statistical significance (P = 0.784).
Postoperative neurological score
Post-operative soft tissue swelling
The swelling of the soft tissue was not very intense and disappeared within the first three post-operative days with the help of hilotherapy. The maximum degree of swelling was reached on the second post-operative day.
Regarding the patient’s satisfaction, which was assessed at the second day after surgery, a statistically significant difference between Piezosurgery® and the conventional saw could be detected (Group 1: 3.1 ± 0.3, Group 2: 1.9 ± 0.2, P = 0.003) Table 1.
The main scope of this clinical, comparative study was to evaluate whether a surgically assisted rapid maxillary expansion using piezotechnology is at least as effective and good as using a conventional oscillating saw.
In the course of the development of orthodontics and maxillofacial surgery different methods for maxillary expansion were tried. When having a closer look at the corresponding literature one sees that no exact indications of how to accomplish a maxillary expansion are defined. The success of a maxillary expansion has been known for centuries, but its efficacy for early treatment cannot yet be rated. As a rule, orthodontists urge an early start of therapy because at that time the growing activity of the bones can be utilized. Nevertheless, there are serious limits, such as orthognathic deseases beyond which an orthodontist should contact a maxillofacial surgeon. In reality, this happens rarely. A similar issue arises when orthodontists do not recommend a maxillary expansion because it requires surgery and, therefore, contains some risk, but instead prefer the extraction of healthy premolars.
The control of the laboratory blood values was conducted by the author himself, as well as the interventions applying piezosurgery. The conventional and piezoelectrical surgeries were accomplished by another surgeon. Therefore, a short learning curve was necessary to get used to the new cutting device. Since piezosurgery needs some time of familiarization, the first five piezosurgery patients were not included in this study.
Due to the exact incision and accurate osteotomy, no arterial vessels were injured and, consequently, no major bleeding occurred. Both patient groups lost approximately the same amount of blood, but since piezosurgery took an average of 10 minutes longer, for Group 2 the loss of blood per unit of time was less than for Group 1. For the author and his team the post-operative condition of the patients that were operated on with the piezo-saw was unambiguous. Some of them were physically more active after a much shorter period of time than Group 1 patients and could, therefore, leave the hospital earlier. However, some patients of Group 2 required a higher degree of pain medication and had to stay another day. A better method for patient selection would have been selecting according to age and gender. This way it would have been easier to distinguish between objective and subjective pain of the different age and gender groups. Due to the random assignment, Group 2 included more females and a larger number of young people. Presumably this is the reason why Group 2 displayed more pain and more circulatory problems. This assumption can be affirmed by the fact that within Group 1 the younger and female patients exhibited more pain and circulatory problems as well. Nevertheless, this study worked with people of all ages to guarantee that different anatomical bone structures with diverse bone qualities had to undergo osteotomy.
The spectrum of maxillofacial procedures has broadened significantly during the last centuries and the interventions have become more and more complex and invasive [27–29]. This is why it became and still is more important to develop operating techniques that are more precise and gentle [30, 31]. Piezoelectric surgery could represent one of those techniques. The results of many piezoelectric osteotomies according to Vercelotti et al. [15–17, 32, 33], Borman et al., Robiony et al.  and many more encourage the author to apply the piezoelectric osteotomy device for even more indications. It is histologically provable that in the long run wound healing is better with this procedure than after surgery with an oscillating saw . Piezosurgery is not a method for fast surgeries, but it is suitable for sensitive and non-traumatic operation procedures.
Neither the laboratory values nor the duration of inpatient stay differed significantly between the two groups. Looking at the height of the blood level in the paranasal sinus piezosurgery showed a statistically significant difference. In conclusion, the question that was posed at the beginning of this paper can be given a positive answer. It is possible to conduct a surgically assisted rapid maxillary expansion with the help of an ultrasonic-saw, piezosurgery, which preserves the mucous membrane of the maxilla and is at least as effective and good as the conventional method. The high performance in terms of frequency and power of the piezosurgical device allow it to be used without the aid of any other osteotome, and with the same atraumatic effect on critical vascular structures. The very low amount of bleeding observed during surgery, lack of damage to the main vessels and reduction of postoperative consequences (hematomas, swelling) for patients were striking.
It is possible to do a surgically assisted palatal expansion of the maxilla by using an ultrasonic bone-cutting device and thus protect the mucous membranes of the maxilla. This method of piezoelectric surgery with all its advantages is going to replace many conventional operating procedures in oral and maxillofacial surgeries.
Approval for the study was obtained from the relevant ethics committee at the University of Muenster, Germany (CIS 2007-237-M). In addition, positive written consent was obtained from each subject who participated in the study.
Rapid maxillary expansion
Surgically assisted rapid maxillary expansion
Slow maxillary expansion
The article processing charges are funded by the Deutsche Forschungsgemeinschaft (DFG), “Open Acess Publizieren”.
- Codivilla A: On the means of lengthening, in the lower limbs, the muscles and tissues which are shortened through deformity. Am J Orthop Surg. 1905, 2: 353.Google Scholar
- Ilizarov GA: Clinical applications of the tension–stress effect for limb lengthening. Clin Orthop. 1990, 250: 8-26.PubMedGoogle Scholar
- Mommaerts MY: Transpalatal distraction as a method of maxillary expansion. Br J Oral Maxillofac Surg. 1999, 37: 268-272. 10.1054/bjom.1999.0127.View ArticlePubMedGoogle Scholar
- Silverstein K, Quinn PD: Surgically-assisted rapid palatal expansion for management of transverse maxillary deficiency. J Oral Maxillofac Surg. 1997, 55: 725-727. 10.1016/S0278-2391(97)90587-5.View ArticlePubMedGoogle Scholar
- Saatci P, Filiz Y: The effect of premolar extractions on tooth-size discrepancy. Am J Orthod Dentofac Orthopedics. 1997, 111 (4): 428-434. 10.1016/S0889-5406(97)80025-X.View ArticleGoogle Scholar
- Isaacson RJ, Murphey TD: Some effects of rapid maxillary expansion. Part I: design of the force measuring system. Angle Orthod. 1956, 34: 256.Google Scholar
- Weise W: Beitrag zur Gaumennahterweiterung. Zahnärztl.Welt/Reform. 1963, 750-758.Google Scholar
- Kraut RA: Surgically assisted rapid maxillary expansion by opening the midpalatal suture. J Oral Maxillofac Surg. 1984, 42: 651-655. 10.1016/0278-2391(84)90207-6.View ArticlePubMedGoogle Scholar
- Messer EJ, Bollinger TE, Keller JJ: Surgical-mechanical maxillary expansion. Quintessence Int. 1979, 10: 13-16.PubMedGoogle Scholar
- Pogrel MA, Kaban LB, Vargervik K, Baumrind S: Surgically assisted rapid maxillary expansion in adults. Int J Adult OrthodonOrthognath Surg. 1992, 7: 37-41.Google Scholar
- Mehra P, Cottrell DA, Caiazzo A, Lincoln R: Life-threatening, delayed epistaxis after surgically assisted rapid palatal expansion: a case report. J Oral Maxillofac Surg. 1999, 57: 201-204. 10.1016/S0278-2391(99)90241-0. [comment in 57:1277]View ArticlePubMedGoogle Scholar
- Neubert J, Somsiri S, Howaldt HP, Bitter K: Die operative Gaumennahterweiterung durch eine modifizierte Le Fort I-Osteotomie. Dtsch Z Mund Kiefer Gesichtschir. 1989, 13: 5764.Google Scholar
- Vercellotti T, Crovace A, Palermo A, Molfetta A: The piezoelectric osteotomy in orthopedics: clinical and histological evaluations (pilot study in animals). Mediterranean J Surg Med. 2001, 9: 89-95.Google Scholar
- Vercellotti T, Majzoub Z, Sabbini E, Valente G, Trisi P, Cordioli G/Università degli Studi di Padova: Comparative histological and clinical evaluation of cutting efficiency: piezoelectric scalpel vs. surgical saw and drill in rabbit calvaria osteotomies. Mediterranean J Surg Med. 2001, 9: 89-95.Google Scholar
- Vercellotti T: Bone Regeneration Techniques in Implantology. Horizontal and Vertical Bone Defect Regeneration. Using Piezosurgery. In 31 st Annual Meeting - USC Periodontal & Implant Symposium. 2006, LA (USA): Marriott - Los Angeles Airport, A29.Google Scholar
- Vercellotti T: La Chirurgie osseuse pièzoèlectrique en implantologi. 2005, Paris: SFBSI - Octobre 22, 4eme ACongrès NationalGoogle Scholar
- Vercellotti T: 15th Annual Implant Symposium. Piezoelectric Bone Surgery in Implantology. 2004, New York: New York University College of DentistryGoogle Scholar
- Vercellotti T, Nevins ML, Kim DM, Nevins M, Wada K, Schenk RK, Fiorellini JP: Osseous response following resective therapy with piezosurgery. Int J Periodontics Restorative Dent. 2005, 25: 543-549.PubMedGoogle Scholar
- Eggers G, Klein J, Blank J, Hassfeld S: Piezosurgery: an ultrasound device for cutting bone and its use and limitations in maxillofacial surgery. Br J Oral Maxillofac Surg. 2004, 42: 451-453.View ArticlePubMedGoogle Scholar
- Schaller BJ, Gruber R, Merten HA, Kruschat T, Schliephake H, Buchfelder M, Ludwig HC: Piezoelectirc bone surgery: a revolutionary technique for minimally invasive surgery in cranial base and spinal surgery? Technical note. Neurosurgery. 2005, 57 (4 supple): E410-discussion, E410PubMedGoogle Scholar
- Vercellotti T: Technological characteristics and clinical indications of piezoelectric bone surgery. Minerva Stomatol. 2004, 53: 207-214.PubMedGoogle Scholar
- Hoigne DJ, Stubinger S, Von Kaenel O, Shamdasani S, Hasenboehler P: Piezoelectic osteotomy in hand surgery: first experiences with a new technique. BMC Musculoskelet Disord. 2006, 7: 36-10.1186/1471-2474-7-36.View ArticlePubMedPubMed CentralGoogle Scholar
- Preti G, Martinasso G, Peirone B, Navone R, Manzella C, Muzio G, Russo C, Canuto RA, Schierano G: Cytokines and growth factors involved in the osseointegration of oral titanium implants positioned using piezoelectric bone surgery versus a drill technique: a pilot study in minipigs. J Periodontol. 2007, 78: 716-722. 10.1902/jop.2007.060285.View ArticlePubMedGoogle Scholar
- Bays RA, Greco JG: Surgically assisted rapid palatal expansion: an outpatient technique with long term stability. J Oral Maxillofac Surg. 1992, 50: 110-113. 10.1016/0278-2391(92)90352-Z.View ArticlePubMedGoogle Scholar
- Lanigan DT, Mintz SM: Complications of Surgically assisted rapid palatal expansion: review of the literature and report of a case. J Oral Maxillofac Surg. 2002, 60: 104-110. 10.1053/joms.2002.29087.View ArticlePubMedGoogle Scholar
- Ueki K, Nakagawa K, Marukawa K, Yamamoto E: Le Fort I osteotomy using an ultrasonic bone curette to fracture the pterygoid plates. J Cranio Maxillofac Surg. 2004, 32: 381-386. 10.1016/j.jcms.2004.06.005.View ArticleGoogle Scholar
- Bell WH, Epker BN: Surgical orthodontic expansion of the maxilla. Am J Orthod. 1976, 70: 517-528. 10.1016/0002-9416(76)90276-1.View ArticlePubMedGoogle Scholar
- Bell RA: A review of maxillary expansion in relation to rate of expansion and patients’ age. Am J Orthod. 1982, 81: 32-37. 10.1016/0002-9416(82)90285-8.View ArticlePubMedGoogle Scholar
- Balters W: Allgemeines zur Atmung und zur Atmungsstörung. Fortschr Kieferorthop. 1954, 15: 193-200. 10.1007/BF02169946.View ArticleGoogle Scholar
- Basciftci FA, Mutlu N, Karaman AI, Malkoc S, Kucukkolbasi H: Does the timing and method of rapid maxillary expansion have an effect on the changes in nasal dimensions?. Angle Orthod. 2002, 72: 118-123.PubMedGoogle Scholar
- Bell WH, Turvey TA: Surgical correction of posterior crossbite. J Oral Surg. 1974, 32: 811.PubMedGoogle Scholar
- Vercellotti T, Podestá A: Piezoelectric surgery in orthodontic therapy: a monocortical tooth distraction technique. Int J Periodontics Restorative Dent. 2001, 21: 561-567.PubMedGoogle Scholar
- Vercellotti T, Obermair G: Introduzione alla Chirurgia Piezoelettrica®. Implantol Dent. 2005, 2: 78-82.Google Scholar
- Bormann K-H, Schramm A, Schmelzeisen R: Piezoelektrische Osteotomie des Nervus alveolaris inferior; Vortrag auf dem 8. 2004, Bad Hofgastein: Kongress der Österreichischen Gesellschaft für Mund-,Kiefer und GesichtschirurgieGoogle Scholar
- Robiony M, Polini F, Costa F, Vercellotti T, Politi M: Piezoelectric bone cutting in multipiece maxillary osteotomies. Technical Note. J Oral Maxillofac Surg. 2004, 62: 6.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.